Press this button to turn On/Off the monitor.
Flash when alarms happen.
Test paper port
Insert the test paper into the port to test Blood glucose.
Type-C input connector
Link to the external power rated 5V/2A.
It turns green when the external power is connected with
Display measurement data and touch screen for operation.
NIBP cuff connector
Link to the NIBP cuff.
SpO₂ sensor connector
Link to the SpO2 sensor.
ECG cable connector
Link to the ECG cable.
TEMP probe connector
Link to the temperature probe.
Play Alarms voice and key tone hrough these holes.
♦ On: When “On” is selected, a detected pacemaker signal is indicated
by a "|" symbol above the ECG waveform.
♦ Off: When “Off” is selected, the pacemaker analysis is disabled.
x0.25, x0.5, x1, x2
Adjust the amplitude of the ECG waveforms. A “1mV” scale is
displayed at the right side of each ECG waveform, the height of the
1mV bar is directly proportional to the ECG waveform amplitude.
3 Lead,5 Lead
Determines whether filter or not.
♦ On: When “On” is selected, the monitor protects the signals from
the noise generated by the power line.
♦ Off: When “Off” is selected, there is no filtering for outside signals.
Diagnostic, Monitor, Surgery
The filtering enables clearer and more detailed waveforms. There are
three filter methods for selection.
♦ Diagnostic: The monitor displays the ECG waveforms without filter;
♦ Monitor: It effectively filters the artifacts that might cause false
♦ Surgery: This filter is used to reduce the artifacts and interference
from electrosurgery equipment.
The selected filter is applied to both channels, but the filter label is
merely displayed above the first ECG waveform.
HR comes from “ECG”, “SpO₂”, “Auto”.
♦ ECG: HR is always calculated from ECG.
♦ SpO2: If the ECG signal is seriously interfered, you can select SpO2,
which means PR will be derived from PLETH waveform.
♦ Auto: The priorities for heart rate calculation are:ECG, SpO2.
♦ On:the monitor will have a self-calibration for ECG waves.
♦ Off:self-calibration will stop.
High, Medium, Low
Off, 10s, 15s, 20s, 25s, 30s, 35s, 40s
Apnea Time Determines whether the patient’s cessation of breath is
an apnea event.
x0.25, x0.5, x1, x2, x4
6.25 mm/s, 12.5 mm/s, 25 mm/s
Aanual, Auto, Continuous
♦ Manual: Measurement on demand;
♦ Auto: Continually repeated measurements at set intervals;
♦ Continuous : A continuous measurement for five minutes.
At any time in NIBP measurement, presseing the NIBP button in the
button area will stop the current NIBP measurement.
Off, 1min, 2min, 3min, 4min, 5min, 10min, 15min, 30min, 60min,
90min, 2hour, 3hour, 4hour, 8hour
Period is adjustable only when mode is “Auto”. Different time intervals
can be selected to determine the interval between automatic
It is used to detect whether the air path leaks. When the cuff is
connected, this function can be used to start the air path inflation
process, so as to find whether the air path is in good sealing condition.
If the air leakage test is passed, the system will not give any prompt; if
not, there will be corresponding error prompt in the noninvasive blood
pressure parameter area.
Restore the initial settings of the pressure pump.
It depends on the type of temperature probe.