Latex test with result in 2 minutes, high sensitivity and specificity, accurate and reliable
Salmonella latex agglutination testing reagent for for identification of suspicious colonies separated from selective plates .The kit is for professional use.
Product list :
Salmonella latex identification kit
Staphylococcus latex identification kit
E. coli O157 latex identification kit
Legionella latex identification kit
Streptococcus latex identification kit
D Group Streptococcus latex identification kit
Listeria latex identification kit
Packing :50 tests / kit
1.Offering Free Sample for Testing .
2.R & D team to provide technical support.
3.Provide packaging, labeling OEM customization.
Sometimes color of dehydrated culture media between batches have subtle differences,would this affect test results?
A1:The source and storage conditions of raw material is different ,so there may be a slight difference in color, which is a normal phenomenon. Products from our company have to undergo a rigorous inspection and sensory biology verification before sale, to ensure that all the indicators characteristics of the product is under enterprise standard to the extent permitted, not affect the test results.
After pouring the liquid medium on the plate , medium seems hard to clot or coagulation time is longer, is it some problem with product quality?
A2: The reason may be:
The required hydration in the preparation process have not been done completely .That is to say distilled water is not boil sufficiently to dissolve agar. Because agar proportion, easily settle in the bottom of the bottle, if not fully shaken after high-temperature sterilization, it will lead to uneven upper agar culture gene content and difficult to set.
Why some colonies of E. coli chromogenic medium color not shown above, but also confirmed through biochemical tests for E. coli (false negative)?
A3: E. CHROMOGENIC medium is based on the principle of specific enzyme of E. coli and design, 94% of E. coli have β- glucuronidase enzyme, with the chromogenic enzyme substrate medium role in the formation of blue-green colonies. And about 4% of the E. coli does not have β- glucuronidase enzymes, including concern O157: H7 Escherichia coli. Therefore, these E. coli can not be displayed on the color characteristic of E. coli chromogenic medium,it is possible of false-negative on the chromogenic medium. However, the traditional medium of the same can not avoid the problem of false negatives, such as: no gas or slow ferment lactose intolerance may also strain 44.5 ℃ false negatives in the conventional medium. For this small part of the E. coli can be detected using other methods.
During anaerobic or micro-aerobic culture, is it necessary to remove gas bag from package? How to use oxygen indicator?
A4: During anaerobic or micro-aerobic culture, using method of gassing agent should be done according to the manufacturer's instructions. Remove the paper bag from the plastic packaging, but do not cut the paper bag. Torn the outer packaging of oxygen indicator then it can be used.